Ni iFixit ni ces informations de réparation ne sont approuvés par ou affiliés au fabricant de ce dispositif médical.
Wiki réalisé avec l'aide d'étudiants
Une formidable équipe d’étudiants de notre programme éducatif a créé ce wiki.
Respironics V60 Troubleshooting
This is a ventilator made by Phillips with a model number of NOCTN96.
Touch Screen Not Responding Properly
The screen of the ventilator doesn’t respond to your touch or is frozen.
Unresponsive/Frozen Touch Screen
If screen is frozen or the screen is not responsive when pressing or pushing buttons. To re-calibrate or to replace it, use “this guide”.
Internal Battery Is Used Up Prematurely/Ventilator Isn’t Powering On
The battery isn’t working or the device won’t power on at all.
If the internal battery fails to work even if it was fully charged, to remove and replace it, use “this guide”.
Sound Is Distorted In the Speaker
Sound comes out weird or not at all when it is supposed to.
If the entire speaker fails to work properly, whether the sound comes out distorted or not at all, remove and replace it using “this guide”.
Top Cover Damage/Cracks
The top cover of the ventilator is cracked or there is some kind of damage done to it.
Cracked Top Cover
If the top cover gets cracked and needs to be fixed, use “this guide” to help you remove and replace it.
Fan is Making a Weird Sound/Fan is not Operating
While using the ventilator, it makes a weird noise or the fan is not operating.
Cooling fan is not functioning
If the cooling fan is making a weird noise or is not operating at all, to remove and replace it, use “this guide”.
The Air Inlet Filter Is Dirty
When checking the filter, it looks dirty.
Air Inlet Filter Collected Too Much Lint or Dust
While using the ventilator, you should be periodically checking to ensure that the filter does not collect too much stuff (such as lint or dust) so it can properly do its job. In the case that the filter collected too much lint or dust, use “this guide” to remove and replace the filter.